I have since the end of March 2012, a sensor in operation. At my place there is an official measuring station at which I can compare the measurements. Since the beginning, the sensor shows too high values.
According to my observations, the deviation is between +0.028 and +0.035 µS/h (average: 0,032 µS/h). The high values of the sensor is almost always above the limit for my region (= 0.120 µS/h). Changes in the values up or down are almost identical (say 95%) with the official measuring station.
If you want to have a look, follow the links:
My private station: http://www.wetter.debastiani.de/html/radioaktivitat.html
Official station: http://odlinfo.bfs.de/cvdata/082260960.php
I tested the sensor already at various locations, all of them were in the house. Result: no improvement.
As a last try I will hang the sensor out of the window.
Does anyone have any idea what I can try next?
Is there somewhere a way to enter a correction factor?
Am grateful for any help
(Translated with Google)